1. Safety Tips
To protect the skin,it’s not suitable for the following customers:
A. You have a known skin disorder.
B. You have a skin disease such as allergic dermatitics,eczema etc.
C. Who have photosensitization.
D. IF you are experiencing Menstruation,Pregnant or Lacation.
E. If you have sunburn.
F. Black or sunburn skin(Please refer to the following specific color contrast card).
G. Black or fuscescent nevus.
H. Broken skin,a scab or healed wound,cancer or hemangioma.
I. Your eyes and any artificial aesthetic,prosthetic or plastic area.
2. Product Introduction
2.1 Packing list:
IPL machine,Power adapter,Instruction manual(Basic set)
2.2 Structure Description
①Air outlet ②ON/OFF button ③Air inlet ④Light exit ⑤Indicator light of power level
5. FunctionsIntroduction Instructions For Use
5.1 Use a medical alcohol wipe to clean the glass window
5.2 Press the “Power button”For 1 second,it will start and enter into the working state(The green indicator light is on,and you can hear the system is running).
A. You will see thr green indicator light is on after starting up.
B. Then you should move the flash window to your skin,you will see the green light flickering,because there is a human body induction sensor built-in it.And the sensor will be sense the skin when you get close to it.
C. The green light indicator must be flickering,otherwise the device will not work.IF you find that the green light is not flickering,bring it closer to your skin until it does flicker.
D. Click the power button to select the suitable intensity level(5 levels with blue indicatoe lights）The diagram below shows intensity level from low to high level.Any adjustment can be made in the sequence shown with a click of the power button.
Intensity Intensity Intensity Intensity Intensity Intensity
Level 1 Level 2 Level 3 Level 4 Level 5 Level A
E. Once the intensity level is set,move the flash window to the skin which can cover the flash window completely,you can see the power indicator lamp flickering and then click the flash button,then it will flash out.After every flash operation the indicator light will be turned off to charge the power,after 1-3 seconds the light will return to indicating that it is ready for use again.
For perfect results,it is suggested the process of hair removel is repeated 2-3 times and ensuring all the body parts should not be repeated and omitted.
F. Press the power button 3 seconds to shut down it.
G. When finished using,please clean the flash window using the suggested medical alcohol wet-wipe(After frequent use there will be dew white spot inside the lens,this will not infect the normal use and service life of the instrument.